We need your consent to use the individual data so that you can see information about your interests, among other things. Click "OK" to give your consent.
Standard Test Method for Capacity of Mixed Bed Ion Exchange CartridgesTranslate name
STANDARD published on 1.6.2017
Designation standards: ASTM D7513-17
Publication date standards: 1.6.2017
The number of pages: 4
Approximate weight : 12 g (0.03 lbs)
Country: American technical standard
Category: Technical standards ASTM
cartridge capacity, conductivity, ion exchange, mixed bed cartridge,, ICS Number Code 71.100.40 (Surface active agents)
|Significance and Use|
5.1 This test method can be used to evaluate unused mixed bed ion exchange cartridges for conformance to specifications.
5.2 This test method provides for the calculation of capacity in terms of the volume of water treated to a conductivity end point.
5.3 The test method as written assumes that the ion exchange resins in the cartridge are either partially or fully converted to the H+ or OH– form. Regeneration of the resins is not part of this method.
5.4 This test method provides for the calculation of capacity on a cartridge basis.
5.5 This test method may be used to test different size mixed bed resin cartridges. The flow rate of test water and the frequency of sampling are varied to compensate for the approximate volume of resin in the test cartridge.
1.1 This test method covers the determination of the performance of mixed bed ion exchange resin cartridges in the active form when used for deionization. The test can be used to determine the initial capacity of unused cartridges or the remaining capacity of used cartridges. In this case performance is defined as ion exchange capacity (or throughput) to two defined endpoints. The method does not measure organics and does not attempt to determine the ultimate water quality attainable by the cartridge.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use
1.4 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
|2. Referenced Documents|
Do you want to make sure you use only the valid technical standards?
We can offer you a solution which will provide you a monthly overview concerning the updating of standards which you use.
Would you like to know more? Look at this page.
Latest update: 2022-12-02 (Number of items: 2 643 927)
© Copyright 2022 NORMSERVIS s.r.o.