We need your consent to use the individual data so that you can see information about your interests, among other things. Click "OK" to give your consent.
Standard Test Method for Acid Number of Crude Oils and Petroleum Products by Catalytic Thermometric Titration (Includes all amendments and changes 7/9/2018).Translate name
STANDARD published on 1.7.2017
Designation standards: ASTM D8045-17e1
Publication date standards: 1.7.2017
The number of pages: 7
Approximate weight : 21 g (0.05 lbs)
Country: American technical standard
Category: Technical standards ASTM
acid number, corrosion, crude oil, naphthenic acids, thermometric titration,, ICS Number Code 75.040 (Crude petroleum), 75.080 (Petroleum products in general)
|Significance and Use|
5.1 Crude oils and oil sands bitumen contain naturally occurring acidic species. Acidity of crude oil has been implicated in corrosion of distribution and process systems. The relative amount of these materials can be determined by titrating with bases. The acid number is a measure of this amount of acidic substance in the oil under the conditions of the test.
5.2 Acid number of crude and distilled petroleum fractions has been measured by Test Method . Test Method was developed for the analysis of lubricants and biodiesel. The titration solvent used in Test Method does not properly address dissolving difficult samples such as crude oil, bitumen, and high wax samples addressed in this test method. Refer to .
5.3 Test Method is also not applicable to measuring acidity of crudes and highly colored samples because the indicator is not visible or it is difficult to discern a color change to detect the end point of the titration.
1.1 This test method covers the determination of acidic components in crude oil and petroleum products including waxes, bitumen, base stocks, and asphalts that are soluble in mixtures of xylenes and propan-2-ol. It is applicable for the determination of acids whose dissociation constants in water are larger than 10–9; extremely weak acids whose dissociation constants are smaller than 10–9 do not interfere. The values obtained by this test method may not be numerically equivalent to other acid value measurements. The range of KOH acid numbers included in the precision statement is 0.1 mg/g to 16 mg/g.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. Some specific hazards statements are given in Section on Safety Precautions.
1.4 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
|2. Referenced Documents|
Latest update: 2022-11-27 (Number of items: 2 642 910)
© Copyright 2022 NORMSERVIS s.r.o.