Standard Test Method for Elemental Analysis of Soil and Solid Waste by Monochromatic Energy Dispersive X-ray Fluorescence Spectrometry Using Multiple Monochromatic Excitation BeamsTranslate name
STANDARD published on 1.11.2016
Designation standards: ASTM D8064-16
Publication date standards: 1.11.2016
The number of pages: 12
Approximate weight : 36 g (0.08 lbs)
Country: American technical standard
Category: Technical standards ASTM
arsenic, As, cadmium, Cd, chromium, Cd, EDXRF, fundamental parameters, FP, HDXRF, heavy metals, Hg, lead, mercury, MMB-EDXRF, Multiple Monochromatic Beam EDXRF, Ni, nickel, Pb, soil, solid waste,, ICS Number Code 13.030.10 (Solid wastes), 13.080.05 (Examination of soils in general)
|Significance and Use|
5.1 Elemental species such as Cr, Ni, As, Cd, Hg, and Pb are widely used in many industrial processes. These elements have been identified in many former industrial sites driving the need for a quick, easy method for testing on-site at trace levels in soil and solid waste matrices.
5.2 This method may be used for quantitative determinations of Cr, Ni, As, Cd, Hg, and Pb in soil matrices and solid waste. Typical test time is 90 seconds to 15 minutes.
1.1 This test method is based upon energy-dispersive X-ray Fluorescence (EDXRF) spectrometry using multiple monochromatic excitation beams for detection and quantification of selected heavy metal elements in soil and related solid waste.
1.2 This test method is also known as High Definition X-ray Fluorescence (HDXRF) or Multiple Monochromatic Beam EDXRF (MMB-EDXRF).
1.3 This test method is applicable to various soil matrices for the determination of Cr, Ni, As, Cd, Hg, and Pb in the range of 1 to 5000 mg/kg, as specified in and determined by a ruggedness study using representative samples. The limit of detection (LOD) for each element is listed in . The LOD is estimated by measuring a SiO1.4 This test method is applicable to other elements: Sb, Cu, Se, Ag, Tl, Zn, Ba, Au, Co, V, Fe, Mn, Mo, K, Rb, Sn, Sr, and Ti.
1.5 X-ray Nomenclature—This standard names X-ray lines using the Siegbahn convention.1.6 Units—The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.7 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
|2. Referenced Documents|
Do you want to make sure you use only the valid technical standards?
We can offer you a solution which will provide you a monthly overview concerning the updating of standards which you use.
Would you like to know more? Look at this page.
Latest update: 2021-04-15 (Number of items: 2 440 988)
© Copyright 2021 NORMSERVIS s.r.o.