Standard Test Method for Determination of Density and Relative Density of Asphalt, Semi-Solid Bituminous Materials, and Soft-Tar Pitch by Use of a Digital Density Meter (U-Tube)Translate name
STANDARD published on 1.7.2018
Designation standards: ASTM D8188-18
Publication date standards: 1.7.2018
The number of pages: 4
Approximate weight : 12 g (0.03 lbs)
Country: American technical standard
Category: Technical standards ASTM
asphalt, asphalt binder, asphalt emulsion, density, density analyzer, digital U-tube density meter, relative density, soft-tar pitch, specific gravity, U-tube,, ICS Number Code 91.100.50 (Binders. Sealing materials), 93.080.20 (Road construction materials)
|Significance and Use|
5.1 Values of density are used for converting volumes to units of mass and for correcting measured volumes at the measured temperature to a standard temperature using Practice .
Note 2: The quality of the results produced by this standard are dependent on the competence of the personnel performing the procedure and the capability, calibration, and maintenance of the equipment used. Agencies that meet the criteria of Specification are generally considered capable of competent and objective testing, sampling, inspection, etc. Users of this standard are cautioned that compliance with Specification alone does not completely ensure reliable results. Reliable results depend on many factors; following the suggestions of Specification or some similar acceptable guideline provides a means of evaluating and controlling some of those factors.
1.1 This test method covers the determination of the relative density and density of semi-solid asphalt materials, asphalt binders, asphalt cements, and soft-tar pitches by use of a digital density meter (U-tube).
Note 1: Alternate methods for determining the density of semi-solid and solid asphalt materials and asphalt binders include Test Methods , , and .
1.2 The values stated in SI units are to be regarded as standard. The values given in parentheses are for information only. No other units of measurement are included in this standard, with the exception of temperature measurements. Units provided in degrees Fahrenheit are for reference purposes only. Temperatures given in degrees Celsius are to be considered standard.
1.3 The between-laboratory reproducibility of this test method is being determined and will be available on or before June 2021. Therefore, this standard should not be used for acceptance or rejection of a material for purchasing purposes.
1.4 This test method should not be applied to petroleum distillates other than asphalt and asphalt binders. For the determination of density of all other petroleum distillates and viscous oils, use Test Method or , or ISO 12185.
1.5 The text of this standard references notes and footnotes which provide explanatory material. These footnotes (excluding those in tables and figures) shall not be considered as requirements of the standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.7 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
Do you want to be sure about the validity of used regulations?
We offer you a solution so that you could use valid and updated legislative regulations.
Would you like to get more information? Look at this page.
Latest update: 2021-04-12 (Number of items: 2 440 772)
© Copyright 2021 NORMSERVIS s.r.o.