We need your consent to use the individual data so that you can see information about your interests, among other things. Click "OK" to give your consent.
Standard Test Method for Insolubles in Used Lubricating OilsTranslate name
STANDARD published on 1.10.2018
Designation standards: ASTM D893-14(2018)
Publication date standards: 1.10.2018
The number of pages: 5
Approximate weight : 15 g (0.03 lbs)
Country: American technical standard
Category: Technical standards ASTM
insoluble resins, insolubles, lubricating oil, pentane insolubles, toluene insolubles, used,, ICS Number Code 75.100 (Lubricants, industrial oils and related products)
|Significance and Use|
5.1 Pentane insolubles can include oil-insoluble materials and some oil-insoluble resinous matter originating from oil or additive degradation, or both.
5.2 Toluene insoluble materials can come from (5.3 A significant change in pentane insolubles, toluene insolubles (with or without coagulant), and insoluble resins indicates a change in oil which could lead to lubrication system problems.
5.4 Insolubles measured can also assist in evaluating the performance characteristics of a used oil or in determining the cause of equipment failure.
1.1 This test method covers the determination of pentane and toluene insolubles in used lubricating oils.
1.2 Procedure A covers the determination of insolubles without the use of coagulant in the pentane. It provides an indication of the materials that can readily be separated from the oil-solvent mixtures by centrifuging.
1.3 Procedure B covers the determination of insolubles in oils containing detergents and employs a coagulant for both the pentane and toluene insolubles. In addition to the materials separated by using Procedure A, this coagulation procedure separates some finely divided materials that may be suspended in the oil.
Note 1: Results obtained by Procedures A and B should not be compared since they usually give different values. The same procedure should be employed when comparing values obtained periodically on an oil in use or when comparing results determined by two or more laboratories.
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use. 1.6 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
|2. Referenced Documents|
Do you want to make sure you use only the valid technical standards?
We can offer you a solution which will provide you a monthly overview concerning the updating of standards which you use.
Would you like to know more? Look at this page.
Latest update: 2022-12-04 (Number of items: 2 644 225)
© Copyright 2022 NORMSERVIS s.r.o.