Standard Specification for Pipet, Sahli HemoglobinTranslate name
STANDARD published on 1.9.2016
Designation standards: ASTM E1045-00(2016)
Publication date standards: 1.9.2016
The number of pages: 3
Approximate weight : 9 g (0.02 lbs)
Country: American technical standard
Category: Technical standards ASTM
This specification covers reusable Sahli hemoglobin pipets. The pipets shall be made of common spirit bore white back tubing. Pipet design shall be straight and of one-piece construction. Pipet delivery tips shall be made with a gradual or concave taper. Pipet dimensions shall be within tolerance limit indicated in this specification. The pipet shall have markings which include graduation line, volumetric designation, identification, and capacity deviation. Testing methods include capacity test, capacity deviation, pigmentation test, and strain free test.
disposal, glass, tube, Westergren,, ICS Number Code 11.100.30 (Analysis of blood and urine)
1.1 This specification covers reusable pipets calibrated “to contain” 20 cmm of whole blood and used for hemoglobin determinations.
1.2 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
|2. Referenced Documents|
Do you want to make sure you use only the valid technical standards?
We can offer you a solution which will provide you a monthly overview concerning the updating of standards which you use.
Would you like to know more? Look at this page.
Latest update: 2021-04-15 (Number of items: 2 440 988)
© Copyright 2021 NORMSERVIS s.r.o.