Standard Specification for Illuminators Used for Viewing Industrial RadiographsTranslate name
STANDARD published on 1.12.2016
Designation standards: ASTM E1390-12(2016)
Publication date standards: 1.12.2016
The number of pages: 3
Approximate weight : 9 g (0.02 lbs)
Country: American technical standard
Category: Technical standards ASTM
This specification provides the recommended minimum requirements for industrial radiographic illuminators used for viewing industrial radiographic films using transmitted light sources. The function of the illuminator is to provide sufficient illumination and viewing capabilities for the purpose of identification and interpretation of radiographic images. The luminance of the transmitted light shall not be less than 30 candelas per square metre for film densities equal to or less than 2.5 optical density and not less than 10 candelas per square meter for film densities greater than 2.5 optical density. If the illuminator has a diffusing screen, the light shall be sufficiently divergent so that both eyes of the observer receive rays from all parts of the screen. The maximum illuminator luminance, and divergence and diffusion of light shall be tested to meet the requirements prescribed.
illuminator, luminance, radiograph,, ICS Number Code 19.100 (Non-destructive testing)
1.1 The function of the illuminator is to provide sufficient illumination and viewing capabilities for the purpose of identification and interpretation of radiographic images. This specification provides the recommended minimum requirements for Industrial Radiographic Illuminators used for viewing industrial radiographic films using transmitted light sources.
1.2 The illuminator has to ensure the same safety for personnel, or users of any electric apparatus, as specified by electrical standards applicable in the country in which the illuminator is used.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
1.4 Values stated in SI units are to be regarded as the standard. The values given in parentheses are provided for information only.
|2. Referenced Documents|
Do you want to make sure you use only the valid technical standards?
We can offer you a solution which will provide you a monthly overview concerning the updating of standards which you use.
Would you like to know more? Look at this page.
Latest update: 2021-04-12 (Number of items: 2 440 772)
© Copyright 2021 NORMSERVIS s.r.o.