Standard Practice for Obtaining Bispectral Photometric Data for Evaluation of Fluorescent ColorTranslate name
STANDARD published on 1.11.2017
Designation standards: ASTM E2153-01(2017)
Publication date standards: 1.11.2017
The number of pages: 10
Approximate weight : 30 g (0.07 lbs)
Country: American technical standard
Category: Technical standards ASTM
bispectral, bispectrometer, calibration, color, colorimetry, Donaldson matrix, Donaldson radiance factor, fluorescence, instrumental measurement, luminescence, radiance factor, spectrophotometry,, ICS Number Code 35.240.99 (IT applications in other fields)
|Significance and Use|
5.1 The bispectral or two-monochromator method is the definitive method for the determination of the general (illuminant-independent) radiation-transfer properties of fluorescent specimens 5.2 This practice provides a procedure for selecting the operating parameters of bispectrometers used for providing data of the desired precision. It also provides for instrument calibration by means of material standards, and for selection of suitable specimens for obtaining precision in the measurements.
1.1 This practice addresses the instrumental measurement requirements, calibration procedures, and material standards needed for obtaining precise bispectral photometric data for computing the colors of fluorescent specimens.
1.2 This practice lists the parameters that must be specified when bispectral photometric measurements are required in specific methods, practices, or specifications.
1.3 This practice applies specifically to bispectrometers, which produce photometrically quantitative bispectral data as output, useful for the characterization of appearance, as opposed to spectrofluorimeters, which produce instrument-dependent bispectral photometric data as output, useful for the purpose of chemical analysis.
1.4 The scope of this practice is limited to the discussion of object-color measurement under reflection geometries; it does not include provisions for the analogous characterization of specimens under transmission geometries.
1.5 This standard may involve hazardous materials, operations, and equipment. This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.6 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
|2. Referenced Documents|
Latest update: 2021-04-20 (Number of items: 2 442 933)
© Copyright 2021 NORMSERVIS s.r.o.