Standard Test Method for Heat of Reaction of Thermally Reactive Materials by Differential Scanning CalorimetryTranslate name
STANDARD published on 1.4.2018
Designation standards: ASTM E2160-04(2018)
Publication date standards: 1.4.2018
The number of pages: 5
Approximate weight : 15 g (0.03 lbs)
Country: American technical standard
Category: Technical standards ASTM
degree of cure, differential scanning calorimetry, hazard potential, heat of reaction, thermal analysis, thermoset,, ICS Number Code 71.040.40 (Chemical analysis)
|Significance and Use|
5.1 This test method is useful in determining the extrapolated onset temperature, the peak heat flow temperature and the heat of reaction of a material. Any onset temperature determined by this test method is not valid for use as the sole information used for determination of storage or processing conditions.
5.2 This test method is useful in determining the fraction of a reaction that has been completed in a sample prior to testing. This fraction of reaction that has been completed can be a measure of the degree of cure of a thermally reactive polymer or can be a measure of decomposition of a thermally reactive material upon aging.
5.3 The heat of reaction values may be used in Practice to determine hazard potential figures-of-merit Explosion Potential and Shock Sensitivity.
5.4 This test method may be used in research, process control, quality assurance, and specification acceptance.
1.1 This test method determines the exothermic heat of reaction of thermally reactive chemicals or chemical mixtures, using milligram specimen sizes, by differential scanning calorimetry. Such reactive materials may include thermally unstable or thermoset materials.
1.2 This test method also determines the extrapolated onset temperature and peak heat flow temperature for the exothermic reaction.
1.3 This test method may be performed on solids, liquids or slurries.
1.4 The applicable temperature range of this test method is 25 to 600°C.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 There is no ISO method equivalent to this standard.
1.7 This standard is related to Test Method and to NAS 1613, but provides additional information.
1.8 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety, health, and environmental practices and determine the applicability of regulatory limitations prior to use.
1.9 This international standard was developed in accordance with internationally recognized principles on standardization established in the Decision on Principles for the Development of International Standards, Guides and Recommendations issued by the World Trade Organization Technical Barriers to Trade (TBT) Committee.
|2. Referenced Documents|
Latest update: 2021-04-18 (Number of items: 2 441 221)
© Copyright 2021 NORMSERVIS s.r.o.